Optical Current Meter Use in Southern Californiaby Thomas S. Chandler, Hydro.; Water Resour. Div., Geological Survey, U.S. Dept. of the Interior, Garden Grove, CA,
Winchell Smith, (M.ASCE), Hydro.; Water Resources Div., Geological Survey, U.S. Dept. of the Interior, Menlo Park, CA,
Serial Information: Journal of the Hydraulics Division, 1971, Vol. 97, Issue 9, Pg. 1461-1469
Document Type: Journal Paper
The optical current meter is a stroboscopic device with which an operator may visually follow and gage the velocity of particles moving on the water surface. From early prototypes the meter has evolved to a field usable instrument with a low power telescope, an oscillating mirror in the line of sight of the telescope, a variable-speed battery-powered motor, and a tachometer which indicates the rotational speed of the mirror. Surface velocities are computed knowing the distance to the water surface and the angular speed of the oscillating mirror. Surface velocities must be corrected by an appropriate coefficient to represent the mean velocity in the vertical. The optical current meter will find its principal use in measurements of discharge under conditions that preclude use of conventional stream-gaging equipment.
Subject Headings: Light rail transit | Water surface | Fluid velocity | Particle velocity | Oscillations | Telescopes | Rotation | Water discharge measurement | Batteries | North America | California | United States
Services: Buy this book/Buy this article
Return to search